I have a S331E and when I convert my files from .VNA to .DAT, the CW is “ON”. How do I turn it off?
This has happened because the RF Immunity is set for “Low” on the instrument and when you convert them from .VNA to .DAT using Master Software Tools, CW is “ON”. You have to set RF Immunity for “High”. To do this press “Sweep/Setup” and on your right hand side you will see the soft key “RF Immunity”. Press to underscore “High”. Unfortunately, you will have to retake your measurements. This can not be changed after saving.