Can you use a one load if you calibrate with a different load?
You may calibrate the Site Master with one precision load and make measurements with another precision load. Type "N" loads are specified to be better than 42 dB return loss, and 7/16 Din loads are specified to 45 dB. You can never make a measurement better than the quality of the calibration components being used. This is true for ALL test equipment.