When testing with an MT8852B Bluetooth tester, can I perform continuous transmitter tests on un-packetized data?
Yes, you can, by using the "CW measurement" mode. Press the Config hardkey. Then use the arrow keys to select MT8852B, then press Sel. Use the arrow keys to select CW Measurement, then press Sel. This puts the unit into continuous measurement mode. From here you need to select the modulation type and the channel number that the DUT is transmitting on. When everything is set up, press Enter to start measurements. Measurements of power, frequency and modulation are made continuously.